The Effect of a Voice Activity Detector on the Speech Enhancement Performance of the Binaural Multichannel Wiener Filter
© Jasmina Catic et al. 2010
Received: 28 January 2010
Accepted: 5 October 2010
Published: 13 October 2010
A multimicrophone speech enhancement algorithm for binaural hearing aids that preserves interaural time delays was proposed recently. The algorithm is based on multichannel Wiener filtering and relies on a voice activity detector (VAD) for estimation of second-order statistics. Here, the effect of a VAD on the speech enhancement of this algorithm was evaluated using an envelope-based VAD, and the performance was compared to that achieved using an ideal error-free VAD. The performance was considered for stationary directional noise and nonstationary diffuse noise interferers at input SNRs from −10 to +5 dB. Intelligibility-weighted SNR improvements of about 20 dB and 6 dB were found for the directional and diffuse noise, respectively. No large degradations (<1 dB) due to the use of envelope-based VAD were found down to an input SNR of 0 dB for the directional noise and −5 dB for the diffuse noise. At lower input SNRs, the improvement decreased gradually to 15 dB for the directional noise and 3 dB for the diffuse noise.
An increasing number of people suffer from hearing loss, a deficit that can limit them in their interaction with the surrounding world and often severely reduces their quality of life. The most common type of hearing loss is the sensorineural, caused by damage to the inner ear (cochlea). People with sensorineural hearing loss often find it difficult to understand speech in the presence of background noise, even when wearing their hearing aids. Consequences of sensorineural hearing loss vary from one individual to another, but factors that often contribute are reduced audibility, loudness recruitment, reduced frequency selectivity, and reduced temporal resolution. Reduced audibility can be compensated for by a hearing aid through amplification, and loudness recruitment can to some extent be alleviated by compression. However, other contributing factors, such as reduced frequency selectivity or deficits in temporal processing, cannot fully be compensated for by a hearing aid. Even if the hearing loss is located in the cochlea and the higher levels of the auditory system function well, the impaired ear may not be able to pass on the multitude of cues otherwise available in the incoming sound. The internal representation of the signals can then be incomplete and difficult to analyze. It is well known that the intelligibility of speech is tightly connected to the signal-to-noise ratio (SNR) . Thus, the problem of speech intelligibility (SI) in noise can be approached by reducing the noise level. While normal-hearing (NH) people can have a speech reception threshold (SRT; the point where 50% of speech is intelligible) at SNRs in the range of −5 to −10 dB depending on the type of noise , this threshold is typically 5-6 dB higher for hearing-impaired (HI) people . At SNRs comparable to the SRT, a small increase in SNR can improve the intelligibility scores drastically as a 1 dB increase can lead to an improvement of up to 15% . This also implies that even a few dB of elevated SRT in HI listeners can cause substantial problems understanding speech compared to NH listeners. Thus, many HI listeners could benefit from a noise reduction of about 5 dB , depending on the acoustical environment.
The noise reduction techniques used in hearing aids employ either a single-microphone or multiple microphones. Single-microphone techniques have been shown not to improve SI in noise but may improve listening comfort . On the other hand, multimicrophone techniques can exploit the spatial diversity of acoustic sources, ensuring that both temporal and spatial processing can be performed. Several microphone array processing techniques have been shown to improve SI in noise . Particularly, adaptive arrays can in certain conditions reduce impressive amounts of noise. However, while the array benefit in hearing aid applications can be very large in the case of a single noise source in mild reverberation, it reduces considerably when several interfering sources are present or when the environment is reverberant . This is due to the use of small arrays with a limited number of microphones used in hearing aids, which limits the array performance. Nevertheless, as small improvements of a few dB might improve intelligibility significantly, a large SNR improvement is not always necessary.
One potential problem with microphone array processing is that it may affect the hearing aid user's sense of the auditory space. Some studies have shown that the users can localize sounds better when the directionality in their hearing aid is switched off [7, 8]. Preserving the interaural localization cues can have a positive effect on speech intelligibility in complex acoustic environments, as the binaural processor in the auditory system can exploit additional information provided by the two ears. Many HI people are able to take advantage of the low frequency interaural time delays (ITDs) almost as effectively as NH people . Thus, a system that combines noise reduction with preservation of ITDs would be desirable. Such an algorithm has recently been proposed in , as a binaural extension of a multichannel Wiener filter-based speech enhancement algorithm proposed in . In  it was shown theoretically that the binaural version preserves the interaural time delays (ITDs) and interaural level differences (ILDs) of the speech component. It was also shown that the ITDs and ILDs of the noise component are distorted in such a way that they become equal to those of the speech component. Therefore, in , the Binaural Multichannel Wiener Filter (BMWF) algorithm was extended to preserve the ITDs of the noise component. A parameter that can pass a specified amount of noise unprocessed, which is supposed to restore the binaural cues of the noise, was included into the calculation of the Wiener filters. Further, it was shown, using an objective cross-correlation measure, that the ITD cues of the noise component were preserved. The BMWF algorithm has also been evaluated perceptually in terms of lateralization performance  and SRT improvements . The conclusion in  was that correct localization was possible with BMWF processing as long as a small amount of noise was left unprocessed. Regarding the SRT improvements in , it was concluded that the performance was as good as or better than that achieved with an adaptive directional microphone (ADM), a standard directional processing often implemented in hearing aids. The algorithm was developed for arbitrary array geometry with no need for any assumptions about the sound source location or microphone positions, and as such it is robust against microphone gain and phase mismatch, as well as deviations in microphone positions and variation of speaker position . It only relies on the second-order statistics of the speech and noise sources, which allows for an estimation of the desired clean speech component. The algorithm relies on a voice activity detection (VAD) mechanism for estimation of the second-order statistics, that is, the algorithm requires another algorithm that detects time instants in the noisy speech signal where the speech is absent. The studies evaluating the BMWF have used an ideal error-free (perfect) VAD which is not available in practice. Generally, VAD algorithms only work well at moderate-to-high SNRs . It is therefore anticipated that the speech enhancement ability of BMWF in those conditions would not be degraded by using a practical VAD instead of a perfect VAD. However, for hearing aid applications, speech enhancement at low SNRs must be considered for two reasons: (1) the SNRs often found in the environment span the range of −10 to 5 dB and should therefore be included in the evaluation of algorithms for hearing aids  and (2) the SRT point, at which there is highest potential for improving intelligibility, is often found at negative SNRs.
In this study, it is investigated to what extent the noise reduction performance of the BMWF algorithm is affected by a realistic VAD compared to a perfect VAD. The BMWF is connected to an envelope-based VAD and the combined system's noise reduction performance is assessed for different types of noise and different spatial configurations of noise sources. The evaluation is based on objective measures such as the intelligibility-weighted SNR improvement. The paper is organized as follows. Section 2 provides an overview of the Binaural Multichannel Wiener Filter algorithm and the envelope-based VAD. Sections 3 and 4 describe the evaluation methods and present results with stationary directional noise and nonstationary diffuse noise. The nonstationary noise is derived from recordings in a restaurant to approach a real world situation. Section 5 provides a discussion of the potential use of this type of noise reduction processing in hearing aids based on the results obtained in this study.
2. System Model and Algorithms
2.1. System Model
A binaural hearing aid system is considered throughout the present study. There are two microphones on each hearing aid and it is assumed that the aids are linked, such that all four microphone signals are available to a noise reduction algorithm. The processor provides a noise reduced output at each ear.
2.2. Binaural Multichannel Wiener Filter
The noise controlling parameter can take on values between 0 and 1, where puts all effort on noise reduction with no attempt on preservation of localization cues, and puts all effort on preserving localization cues and no noise reduction is performed, that is, there is a trade-off between noise reduction and preservation of localization cues.
The BMWF algorithm uses no information for computation of the filter matrix other than the second-order statistics determined by the VAD. It can be expected that the performance of the BMWF will degrade at some point due to VAD detection errors, leading to incorrect noise estimation. If speech is detected as noise, vectors containing speech samples will be added to the noise data matrix in (5), which leads to cancellation of parts of the speech signal. On the other hand, if too many actual noise samples are detected as speech, less noise vectors are added to the noise data matrix in (5) and a poorer noise estimate is obtained which leads to incorrect noise reduction. Generally, a multichannel Wiener filter can be decomposed into a minimum variance distortionless response MVDR beamformer followed by a (spectral) Wiener postfilter . Therefore, it can also be expected that the speech enhancement strongly depends on the spatial configuration of the noise sources. The adaptive beamformer is mostly effective at suppressing interference comprising fewer sources than the number of microphones, with the noise reduction decreasing fast as the number of noise sources increases. While the beamformer should not modify the target signal, the postfilter can attenuate the target signal, according to the amount of noise present at the output of the beamformer. Hence, as the Wiener postfilter trades off target distortion with noise reduction, the amount of target cancellation is expected to be small in the case of few noise sources, and high for many sources.
2.3. Voice Activity Detector
A 50% overlap is used such that the processing delay is 4 ms. Each frame is Hanning windowed and a 256-point FFT is performed.
Short-term magnitude-squared spectra were calculated. Temporal power envelopes are obtained by summing up the squared spectral components. Moreover, a low- and high-band power envelope are calculated, by summing up the squared spectral components below a cutoff frequency and above . The envelopes of band-limited signals are considered since some noise types have stronger low- (or high-) frequency components. In that case, one of the band-limited envelopes may be less disturbed by the noise and provide more reliable information for speech pause decision. The envelopes are smoothed slightly using a first-order recursive low-pass filter with a release time constant .
The maxima and minima of the signal envelope are obtained by tracking the peaks and valleys of the envelope waveform. This is done with two first-order recursive low-pass filters with attack and release time constants and . The differences between the maxima and minima are calculated to obtain the current dynamic range of the signal.
The decision for a speech pause is based on several requirements regarding the dynamic range of the signal and the current envelope values for the three bands. As the complete decision process is described in , it will not be outlined here, that is, only the general concepts are provided. The criterion for the envelope being close enough to its minimum is determined by the free parameters and and the current dynamic range of the signal. The threshold parameter represents the threshold for determining whether the current dynamic range of the signal is low, medium or high. The parameter can take on values between 0 and 1 and is used in comparisons of whether a fraction ( ) of the current dynamic range is higher than the difference between the current envelope and its minimum. The settings of and determine how strict the requirements for detecting a speech pause are, and they can be adjusted to make the VAD more or less sensitive to detecting speech pauses. By increasing one or both of the parameters, the algorithm will detect more speech pauses, but at the same time, it will also detect more speech periods as noise.
3. Evaluation Setup
The speech enhancement performance of the system was evaluated for SNRs in the range from −10 to +5 dB, as this range is most important for hearing aid applications (see Section 1). Since the performance of microphone arrays strongly depends on the spatial characteristics of the interfering noise, the system was evaluated both in conditions of directional and diffuse noise. Further, two noise types were considered: a stationary noise with low modulation index and a nonstationary noise with strong envelope fluctuations.
3.1. Performance Measures
Several studies on microphone arrays for hearing aids have found good agreement between the weighted SNR improvement and changes in SRTs for normal-hearing individuals [22, 23]. In , a close agreement between the AI weighted directivity index (AI-DI) (in the case of diffuse noise and frontal incidence of target, the approaches the AI-DI) and SRTs for hearing-impaired listeners was reported. Although it can be expected that an improvement in SNR in the frequency regions important for speech intelligibility should improve speech recognition, this measure is not considered as a substitute for speech intelligibility tests with hearing-impaired listeners.
3.2. Reference System
3.3. Experimental Setup
The measurements of speech and noise were carried out in an acoustically highly damped room. The speech and noise sources were recorded separately on behind the ear (BTE) hearing aids with omnidirectional microphones, mounted on a dummy head which was placed in the center of the room. The speech waveform is shown in Figure 2. The 8 seconds long speech segment is a male speaker on BBC news, where an additional speech pause was added to the waveform in the intervals from 3.5 to 4 seconds and 7.5 to 8 seconds. This was done since there are very few natural speech pauses in the newsreader speech, and because the BMWF relies on presence of speech pauses for noise estimation. It is assumed that, in a more natural conversation, several speech pauses would be present in the waveform. The speech was played through a loudspeaker located at 0° azimuth relative to the dummy head. The stationary noise used was speech-shaped noise, which is a steady noise with the same long-term average spectrum as (typical) speech. The noise was recorded at the House Ear Institute in Los Angeles. In order to generate directional noise, this recording was played through a loudspeaker positioned at an azimuth of 90° relative to the dummy head. The nonstationary noise used was diffuse multitalker babble noise. Further recording were made in a restaurant at 8 different locations. These recordings were played from 8 different loudspeakers located in the corners of the room. This artificial diffuse sound field is assumed to mimic a "cocktail party" situation, and was chosen to assess the performance of BMWF combined with envelope-based VAD in a realistic and challenging acoustical environment.
The sampling frequency was 24.414 Hz and the BMWF filter length per channel was 64. The filters in (7) were calculated using the whole signal. The output speech and noise signals were generated by filtering the clean speech and noise signals separately with the obtained filter coefficients. The input SNRs were calculated using the VAD sequence shown in Figure 2 in order to exclude the noise-only samples indicated by zeros from the calculation.
List of parameters used in VAD implementation.
4.1. Speech and Noise Classification
In this section, the speech and noise classification performance of the envelope-based VAD for the three settings of is presented. The percentages of correctly detected samples were calculated for the scenarios described in the experimental setup in Section 3. Hence, the noise reduction and speech cancelation obtained for each scenario in Sections 4.2 and 4.3 can directly be related to this particular classification performance. The correct scores were calculated with respect to the perfect VAD sequence from Figure 2 (Section 3). Note that the length of the entire signal was 8 seconds of which about 2 seconds were noise and so the amount of speech and noise is not equal.
4.2. Stationary Directional Noise
The right panel of Figure 5 shows that reducing from 1 to 0.8 (to preserve ITD cues of the noise component) leads to SNR improvement of about 13 dB for all considered SNR conditions when utilizing perfect VAD. This is substantially less than the 20 dB obtained with the setting. However the degradation of noise reduction performance due to employing envelope-based VAD is smaller when the noise estimate is scaled, such that an average gain of 10 dB is found.
Results in the right panel of Figure 6 show that setting reduces the amount of target cancellation by up to 1.5 dB.
4.3. Diffuse and Fluctuating Noise
The detection rates for noise displayed in Figure 3 show that, as the input SNR decreases, the VAD classifies a higher amount of noise as speech. But this is not the only reason for reduced performance. Figure 3 shows that the VAD detection rates are quite similar at and below −5 dB input SNR, yet the SNR improvement decreases. The noise reduction performance does not only depend on the VAD error rates, but also on the quality of the noise estimate and this is especially pronounced at very low SNRs in nonstationary noise. The noncontinuous collection of noise data introduces inaccuracies in the noise correlation matrix since it is estimated only in limited periods of time in the entire signal waveform. Thus, the filter coefficients differ from those that could have been obtained if the speech and noise correlation matrices were estimated at the same time. While the improvement for directional speech-shaped noise in Figure 5 actually increases with decreasing SNR when employing a perfect VAD, this is not the case for diffuse babble noise (Figure 7), where a 1 dB decrease is seen. Therefore, frequent sampling of the fluctuating noise is even more important at lower SNRs.
The right panel of Figure 7 shows that a setting in diffuse noise results only in a very small decrease in SNR improvement (on average 1 dB).
The noise reduction results showed that for stationary directional noise an average SNR improvement of 20 dB (see left panel of Figure 5) can be achieved when using perfect VAD for noise estimation in the BMWF system. The effect of incorporating a realistic VAD for this scenario is minimal (<1 dB) as long as the input SNR is at or above 0 dB. Although noise reduction performance deteriorated with decreasing SNR, a robust gain of about 15 dB is still obtained at −10 dB input SNR. When trading off some noise reduction in order to preserve ITD cues of the noise component (i.e., setting , shown in right panel of Figure 5), an adequate improvement in SNR of 10 dB on average can still be obtained. This means that in such a situation, the user could, in addition to the benefit from auditory release from masking (that also improves speech intelligibility), also benefit from the microphone array processing. While an adequate amount of noise reduction can be obtained for the case of stationary directional interferer, the noise recorded in a restaurant is a more realistic condition that often would be encountered by hearing aid users. In this scenario, a limited amount of noise reduction of about 6 dB was obtained by the BMWF system in the optimal case (i.e., with perfect VAD), as can be seen in Figure 7. Furthermore, the setting reduced the SNR improvement by 1 dB. It could be argued that this reduction is not necessary since in a diffuse noise environment no directional localization cues for the noise are available. In the present study, it was assumed that the hearing aid user does not adjust the setting according to the acoustical environment, but in principle it should be possible that this adjustment is made in the hearing aid according to the acoustical environment with the sound classifiers installed in modern hearing aids.
When using the envelope-based VAD, the performance is not degraded by more than 1 dB down to an input SNR of about −5 dB compared to the optimal case. At this point (for ), the correct classification of speech was about 78% and the correct classification of noise was about 50% (see Figure 3). Thus, it is not necessary for the BMWF system that the VAD shows satisfactory performance (i.e., a low error rate), but rather that the error rate is not excessive (e.g., higher than 50%), and therefore only small effects of VAD are observed in relatively adverse conditions. It should be noted, that even a small weighted SNR improvement of 3–6 dB found for diffuse babble noise can lead to a crucial speech recognition increase, if the improvement is found at SNRs comparable to the SRT. In , for example, sentence intelligibility in different types of noise for hearing-impaired listeners was investigated. The average SRTs for speech-shaped noise and fluctuating noise were −3.3 dB and −2.1 dB, respectively, with improvements in speech recognition of 16 and 11 percent for each 1 dB increase in SNR. This means that for a typical hearing-impaired individual the SNR range of understanding almost nothing to understanding almost everything is −7 to 3 dB for sentences in fluctuating noise. In much of this SNR range (down to −5 dB), the BMWF performance does not degrade much due to VAD errors and an SNR improvement of 5-6 dB is found. Hence, the BMWF with envelope-based VAD might provide a significant improvement in speech recognition of more than 50%.
In very adverse conditions, for example, at −10 dB SNR, which may also be encountered in the environment, the SNR improvement reduced to about 3 dB when using envelope-based VAD for noise estimation, which is comparable to that of a directional microphone. A first-order directional microphone, consisting of two closely spaced microphones has an AI weighted directivity index as measured on KEMAR (which is equivalent to our measure of weighted SNR improvement in diffuse noise) of around 3 dB, for example, [5, 26]. It should be kept in mind that the results regarding reduction in SNR improvement relative to that obtained when employing perfect VAD are limited to the specific VAD used here. The effect of other types of VAD algorithms may be different. In addition to the degraded performance in very adverse conditions, an obvious problem for this system arises if the interference is a single speaker or only a few speakers. In such situations, the temporal fluctuations of the noise interferer are very similar to the target fluctuations and thus, the VAD cannot discriminate between both. In consequence, no significant suppression of the interferers can be achieved.
The purpose of this work was primarily to investigate the effect of a realistic VAD on BMWF, more specifically, to identify the range of SNRs where the VAD has minimal effect on noise reduction performance compared to the case when VAD errors are not taken into account, and to quantify the degradation in performance for the conditions where the VAD has significant influence. The following aspects can be subject to further research. The analysis presented has employed block processing where the statistics of speech and noise were calculated using the entire signal of 8 seconds of which about 2 seconds were noise. It is likely that head movement and movement of noise sources will degrade algorithm performance. In this context, the performance of the algorithm will not only be influenced by the type of adaptation used, but by the filters only being updated during speech pauses. Obviously, this impedes tracking of fast movement, as the filters can be frozen for seconds to the previous scenario. Also, VAD classification errors can lead to slower convergence of the filters. Due to the directional properties of the BMWF, this degradation is more likely to be significant in a simple (directional) noise source setup than if the noise scenario is complex that is, spatially diffuse.
Although it can be expected that an SNR improvement in frequency regions important for speech recognition would result in higher speech recognition, the gains obtained in intelligibility-weighted SNR can only be related to the potential of this system to improve intelligibility. This is particularly critical when individual hearing impairments (e.g., limitation in audibility, spectral resolution, or temporal fine structure processing) are considered. The effect of hearing impairment on speech intelligibility might be addressed by using modifications to the speech-weighted SNR measure such as those proposed in, for example,  for the Articulation Index. However, in order to demonstrate the true benefit of the BMWF system in complex scenarios, speech intelligibility tests with hearing aid users need to be conducted. Also, the quality of the processed speech could be addressed.
- Moore BCJ: Speech processing for the hearing-impaired: successes, failures, and implications for speech mechanisms. Speech Communication 2003, 41(1):81-91. 10.1016/S0167-6393(02)00095-XView ArticleGoogle Scholar
- Festen JM, Plomp R: Effects of fluctuating noise and interfering speech on the speech-reception threshold for impaired and normal hearing. Journal of the Acoustical Society of America 1990, 88(4):1725-1736. 10.1121/1.400247View ArticleGoogle Scholar
- Plomp R: Auditory handicap of hearing impairment and the limited benefit of hearing aids. Journal of the Acoustical Society of America 1978, 63(2):533-549. 10.1121/1.381753View ArticleGoogle Scholar
- Nilsson M, Soli SD, Sullivan JA: Development of the hearing in noise test for the measurement of speech reception thresholds in quiet and in noise. Journal of the Acoustical Society of America 1994, 95(2):1085-1099. 10.1121/1.408469View ArticleGoogle Scholar
- Kates JM: Digital Hearing Aids. Plural Publishing, San Diego, Calif, USA; 2008.Google Scholar
- Bitzer J, Simmer KU, Kammeyer K: Theoretical noise reduction limits of the generalized sidelobe canceller (GSC) for speech enhancement. Proceedings of the IEEE International Conference on Acoustics, Speech, and Signal Processing (ICASSP '99), March 1999 2965-2968.Google Scholar
- Van Den Bogaert T, Klasen TJ, Moonen M, Van Deun L, Wouters J: Horizontal localization with bilateral hearing aids: without is better than with. Journal of the Acoustical Society of America 2006, 119(1):515-526. 10.1121/1.2139653View ArticleGoogle Scholar
- Klasen TJ, Rohrseitz K, Keidsler G: The effect of multi-channel wide dynamic range compression, noise reduction, and the directional microphone on horizontal localization performance in hearing aid wearers. International Journal of Audiology 2006, 45: 563-579. 10.1080/14992020600920804View ArticleGoogle Scholar
- Bronkhorst AW, Plomp R: Binaural speech intelligibility in noise for hearing-impaired listeners. Journal of the Acoustical Society of America 1989, 86(4):1374-1383. 10.1121/1.398697View ArticleGoogle Scholar
- Klasen TJ, Moonen M, Van Den Bogaert T, Wouters J: Preservation of interaural time delay for binaural hearing aids through multi-channel Wiener filtering based noise reduction. Proceedings of the IEEE International Conference on Acoustics, Speech and Signal Processing (ICASSP '05), 2005 3: 29-32.Google Scholar
- Doclo S, Moonen M: GSVD-based optimal filtering for single and multimicrophone speech enhancement. IEEE Transactions on Signal Processing 2002, 50(9):2230-2244. 10.1109/TSP.2002.801937View ArticleGoogle Scholar
- Cornelis B, Doclo S, Van den Bogaert T, Wouters J, Moonen M: Theoretical analysis of binaural multi-microphone noise reduction techinques. IEEE Transactions on Audio, Speech and Language Processing 2010, 18(2):342-355.View ArticleGoogle Scholar
- Klasen TJ, Van den Bogaert T, Moonen M, Wouters J: Binaural noise reduction algorithms for hearing aids that preserve interaural time delay cues. IEEE Transactions on Signal Processing 2007, 55(4):1579-1585.MathSciNetView ArticleGoogle Scholar
- Van Den Bogaert T, Doclo S, Wouters J, Moonen M: The effect of multimicrophone noise reduction systems on sound source localization by users of binaural hearing aids. Journal of the Acoustical Society of America 2008, 124(1):484-497. 10.1121/1.2931962View ArticleGoogle Scholar
- Van Den Bogaert T, Doclo S, Wouters J, Moonen M: Speech enhancement with multichannel Wiener filter techniques in multimicrophone binaural hearing aids. Journal of the Acoustical Society of America 2009, 125(1):360-371. 10.1121/1.3023069View ArticleGoogle Scholar
- Vary P, Martin R: Digital Speech Transmission—Enhancement, Coding and Error Concealment: Chapter 11. John Wiley & Sons, New York, NY, USA; 2006.View ArticleGoogle Scholar
- Ricketts TA: Directional hearing aids: then and now. Journal of Rehabilitation Research and Development 2005, 42(4, supplement 2):133-144.Google Scholar
- Brandstein M, Ward D: Microphone Arrays—Signal Processing Techniques and Applications: Chapter 3. Springer, New York, NY, USA; 2001.Google Scholar
- Marzinzik M, Kollmeier B: Speech pause detection for noise spectrum estimation by tracking power envelope dynamics. IEEE Transactions on Speech and Audio Processing 2002, 10(2):109-118. 10.1109/89.985548View ArticleGoogle Scholar
- Greenberg JE, Peterson PM, Zurek PM: Intelligibility-weighted measures of speech-to-interference ratio and speech system performance. Journal of the Acoustical Society of America 1993, 94(5):3009-3010. 10.1121/1.407334View ArticleGoogle Scholar
- ANSI S3.5-1997 : American National Standard Methods for Calculation of the Speech Intelligibility Index. The Acoustical Society of America, 1997
- Peterson PM, Wei S-M, Rabinowitz WM, Zurek PM: Robustness of an adaptive beamforming method for hearing aids. Acta Oto-Laryngologica 1990, (469, supplement):85-90.
- Hoffman MW, Trine TD, Buckley KM, Van Tasell DJ: Robust adaptive microphone array processing for hearing aids: realistic speech enhancement. Journal of the Acoustical Society of America 1994, 96(2):759-770. 10.1121/1.410313View ArticleGoogle Scholar
- Laugesen S, Schmidtke T: Improving on the speech-in-noise problem with wireless array technology. News from Oticon 2004, 3-23.Google Scholar
- Wagener KC, Brand T: Sentence Intelligibility in noise for listeners with normal hearing and hearing impairment: influence of measurement procedures and masking parameters. International Journal of Audiology 2005, 44(3):144-156. 10.1080/14992020500057517View ArticleGoogle Scholar
- Harnacher V, Chalupper J, Eggers J, Fischer E, Kornagel U, Puder H, Rass U: Signal processing in high-end hearing aids: state of the art, challenges, and future trends. Eurasip Journal on Applied Signal Processing 2005, 2005(18):2915-2929. 10.1155/ASP.2005.2915View ArticleGoogle Scholar
- Pavlovic CV, Studebaker GA, Sherbecoe RL: An articulation index based procedure for predicting the speech recognition performance of hearing-impaired individuals. Journal of the Acoustical Society of America 1986, 80(1):50-57. 10.1121/1.394082View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.